4. Select the Analysis View and extinguish plasma by pressing the Plasma icon.
5. Exit the software
*The program will not shutdown unless the plasma has been turned off.
6. Turn off cooling water
*Do not turn off the cooling water before the plasma. This will cause a cooling error on the next run and could be possibly detrimental to the instrument.